This project has become kinda boring (esp.
w/o pay), so no more observations!
Kepler Star KIC 8462852 Amateur Photometry Monitoring Project
B. L. Gary, Last updated 2017.01.15 UT
This web page records my observations of KIC 8462852 (hereafter
KIC846) in an attempt to measure small amplitude variations on
timescales of hours, days and weeks. A secondary goal is to begin
a multi-year measurement of a suggested gradual decline of
0.34%/year (Montet & Simon, 2016). In support of both goals 25
nearby stars have been calibrated using all-sky photometry.
All-sky photometry for B- and V-band was completed on Sep 25,
involving 25 nearby stars. I've begun the monitoring phase of
observations. If consideration is restricted to observations with my
standard hardware configuration (Meade 14" and SBIG ST-10XME CCD)
KIC846 has remained constant at 11.90 V-mag since September, with a
RMS departure from average of 0.0028 mag. (The observations between
Sep 25 and Oct 31 were made with backup hardware while my Meade, and
later CCD, were being repaired.) There's "a hint" of a 30-day
sinusoidal variation with amplitude 2.7 mmag; I don't believe it
yet. Here is the latest day-timescale light curve:
Latest Status Report
If we discount the observations made with backup telescopes and
CCDs (272 < DOY < 306), and use only data taken with the
M14/ST10 hardware, there is no credible evidence of brightness
change during the past 2 months, except for a very small
sinusoidal-appearing 30-day variation.
This data is made with identical hardware configuration after
telescope and CCD repairs. It is fitted with a model that
includes a steady fade rate (1.4 ± 1.1 %/year), and a sinusoidal
variation (2.7 ± 0.8 mmag semi-amplitude with period of 30
days). Although the formal uncertainties indicate that the
sinusoidal variation is statistically significant (at the
3.6-sigma level), I have insufficient knowledge of systematic
errors to believe this variation.
Links on this Web Page
photometry B- and V-band calibration of KIC846 star field
(linked to another web page)
KIC846 basic info
Finder chart with
calibrated star B-mag and V-mag table
Goals & Plan
My original goal was to detect the suggested fade rate
of 0.34%/year during the 2016 observing season. The suggested fade
rate corresponds to 3.4 mmag/year, or 1.0 mmag every 3.5 months, which is the
length of my observing season (due to the summer monsoon ending
in mid-September). However, I have determined that
systematic errors are present at the level of ~ 3 mmag per
and I do not know what systematic errors may exist during
monthly timescales, so it is unreasonable to expect success
in measuring a 1.0 mmag fade during a 3.5-month interval. Instead,
I will attempt to measure the slow secular fade by comparing
results during two observing seasons.
My original secondary goal was to search for short timescale
variability, and this has become a primary goal for the
current observing season. A few observing sessions will be
long enough to search for hourly variations, but most observing sessions
will be limited to an hour. This should allow for a search of day-to-day
changes, as well as weekly timescale changes. If none are found then
the observations will be viewed as providing an observational
upper-limit to such variability.
The observational strategy is to
improve the calibration of nearby stars so that each observing
session produces a better-calibrated V-mag for that date. The
tighter the star color sensitivity scatter plot, which I use
for establishing an observing session's calibration, the
smaller are the systematic errors for that session's target
V-mag. The APASS magnitudes for this star field appear to be
flawed, in both average value and internal consistency
(star-to-star ratios). I will use all-sky photometry,
involving many Landolt stars, to accomplish this calibration.
In order to know how long each weekly observing session will
have to be for the average target V-mag to have a stochastic
SE smaller than the estimated systematic SE I will conduct a
couple long observing sessions (> 6 hrs) and simulate
having just limited time chunks of the data, with a sampling
chunk lengths. KIC846 is bluer than all but one nearby star to
be used for calibration. This means I have to worry about air
mass effects that differ between the target (KIC846) and the
set of nearby calibrator stars. In order to assess the
usability of high air mass data I will determine a best value
for my "air mass curvature correction," AMC', and demonstrate
the stability of this parameter, and use simulations of long
observing sessions to determine how safe it is to use only
high air mass observations (made late in the observing season)
for monitoring secular fade rate.
I will sue the same hardware configuration for all
observations: Made 14" LX200 GPS telescope, in a dome, with a
x2 focal reducer (designed for this Meade model, A CFW-10 with
an Astrodon V-band filter and an SBIG ST-10XME CCD. All
hardware control is performed using MaxIm DL, via 100-foot
buried conduit cabling between my residence office and the
dome observatory. The unbinned image scale for this system is
0.725 "arc/pixel, and the FOV is 28 x 18 'arc. I will use 2x2
binning for all observations, placing the target at the center
of the image, and autoguiding with the 2nd chip of the
ST-10XME CCD. This will remove drift of the star field with
respect to the pixel field during an observing session, and if
I place the target accurately at the same center location for
each observing session this should reduce the effects of
imperfect flat field calibration.
Basic Info for KIC846
RA/DE = 20:06:15.5,
All-sky photometry: B = 12.493 ± 0.025, V = 11.912 ±
0.025 (B-V = +0.581 ± 0.035)
APASS Mag's: B = 12.360, V = 11.852 (B-V = +0.51), g' =
12.046, r' = 11.697, i' = 11.554
There's a 0.11
mag discrepancy between BV mag's in
Boyajian et al (Table 3) and APASS (article is
There's a 0.23 & 0.21 mag discrepancy between BV
mag's in Boyajian et al (Table 3) and my all-sky V-mag
(article is brighter).
Sessions (Most Recent at Top)
All-sky session, for V-band. Same V-mag as previous ones: V-mag =
Using Meade 14" with ST-10XME CCD, V-band. Finally, a reliable
system! Data quality is better due to large FOV with smooth flat
field and V filter (small Star Color Sensitivity slope).
Switched back to no filter (bec V filter had bubbled surface,
causing bad flat field).
I began using a V-band filter with the Lodestar.
The variations are undoubtedly due to flat field imperfections!
The wiggles near the end must be due to flat field imperfections.
The wiggles near the end must be due to flat field
CCD failure occurred at 3:30 UT!
Fifth C11 observing session.
Fourth C11 observing session.
Third C11 observing session.
Second C11 observation; a slight increase in brightness.
First observation with the Celestron 11" telescope (C11).
Note: On 2016.10.01 the Meade 14"failed to
turn on, so I moved the SBIG ST-10XME to my Celestron 11"
telescope (also in a dome), and have continued observations with
this "back-up" telescope system.
This observing session had 4 goals, or questions to answer: 1) Does
the "star color sensitivity" diagram exhibit the same small MRS
scatter about a slope predicted by the all-sky measurements of the
previous week?, 2) What's the "air mass curvature" coefficient of
the blue target star when calibrated by mostly red stars?, 3) Is
there evidence for smal amplitude, short-period variations during a
long observing session? and 4) How accurately can a 1-hour observing
session measure V_mag?
Based on cal using V mags adjusted on 2016.11.15.
Uses the adjusted V-mags of 2016.11.15 (based on 2016 Nov 10, 12, 14
After minor adjustments of 11 stars for persistent departures
from Star Color Sensitivity plots (based on 3 observing sessions,
Sep 25, Oct 02, Oct 03).
Star color sensitivity relationship for all-sky calibrated nearby
stars, with an offset solution used for setting the calibration
for this date's observing session.
The star color sensitivity relationship has a small slope, as
expected (same as before), and the scatter of 13 mmag about a slope
fit is what I expected, based on the all-sky measurements. This
scatter is ~ 1/2 of what was present when APASS mag's were used.
This 6.6-hr LC employs an AMC of only 3.0 mmag/airmass. Even using
zero produces a LC that is almost indistinguishable to the eye. When
airmass < 2 it doesn't matter whether AMC is zero or 3.5.
Averaging mag's from groups of 5-images yields a RMS scatter of 1.3
mmag. The formal sinusoid solution is amplitude = 1.3 +/- 0.4 mmag,
P = 1.0 hr. This is below my empirical threshold for being real, so
I conclude that during this 6.6-hr observing session there were no
detectable variations (above ~ 2 mmag). For future reference, a
1-hour data segment (above EL 30 deg) should be sufficient to
determine V-mag with a precision (and repeatability?) of ~ 3 mmag.
This SE is dominated by the use of 23 cal stars and their "star
color sensitivity" solution. The best AMC' = +0.003 mag/airmass. The
effect of AMC' departing from zero is not apparent in the LC,
visually, until airmass > 2.
To investigate how long a data chunk must be in order for its
stochastic SE to be smaller than estimated systematic error (~ 3
mmag), I divided the data into 1-hour chunks of data. The SE per
hourly median is 0.9 mmag, so hour-long observations provide an
average V-mag with a stochastic SE that is several times smaller
than estimated systematic uncertainty. Here's a plot of the hourly
medians and 5-image averages.
LC for 5-image averages (blue diamonds) and hourly averages (red
I conclude that 20-minute observations should be adequate for
monitoring KIC846 on a weekly basis in search of a secular fade.
B-band all-sky observations for calibrating the target star field,
involving 8 Landolt star field (25 stars). The observing sequence
was L1-t-L2-t-L3 ... t-L8, where Ln = Landolt star field #n and t =
V-band all-sky observations for calibrating the target star
field, involving 7 star fields.
DISREGARD THIS SECTION; IT'S LEFT-OVER OBSERVATION LIGHT CURVES FROM
AN EARLIER KIC846 MONITORING PROJECT THAT I'VE ABANDONED
2015.11.12, B. Gary, C filter LC, r'-mag = 11.703 ± 0.005
2015.10.24, B. Gary, C filter LC, r'-mag =
11.689 ± 0.003
2015.10.23, B. Gary, C filter LC, r'-mag = ??.??? ± ?.??? (dew problem)
Gary, C filter LC, r'-mag = 11.691 ± 0.005
2015.10.16, B. Gary, C filter LC, r'-mag = 11.697 ± 0.005
2015.10.15, B. Gary, C filter LC, r'-mag =
11.689 ± 0.010
2015.11.12: B. Gary, Meade 14", unfiltered,
calibrated using APASS r'-mags, 1.3 hrs.
2015.10.24: B. Gary, Meade 14", unfiltered, calibrated using
APASS r'-mags. 5.8 hr.
Faint gray trace is a predicted variation based on previous
observing sessions (which doesn't allow for air mass curvature).
2015.10.23: B. Gary, Meade 14", unfiltered, calibrated using
APASS r'-mags. 3.0 hr, but due to dew accumulation on the corrector
plate only the first 2 hrs were useable.
2015.10.20: B. Gary, Meade 14", unfiltered, calibrated using
APASS r'-mags. 0.7 hr.
The small-amplitude sinusoidal variation, with P = 0.8 hr,
appears to be statistically significant (same P as in previous
2015.10.16: B. Gary, Meade 14", unfiltered, calibrated using
APASS r'-mags. 4.2 hrs.
Best 5 ref stars chosen for color similarity to target &
other behavior. The small-amplitude sinusoidal variation, with P =
0.8 hr, appears to be statistically significant.
Square root of "Power Spectrum" showing weird shape.
Finder Image and
Calibrated Star B-mag and V-mag Table
This is a finder chart.
FOV = 26 x 18'arc, north up, east left. KIC846 is in the square.
Here's a smaller FOV finder chart showing 25 stars that have been
calibrated using all-sky photometry.
FOV 14.0 x 9.4 'arc showing KIC846 (red square) and 25 calibrated
reference stars. North up, east left.
The table below is a listing of my all-sky photometry B- and V-
All-sky photometry of KIC846 and 25 nearby stars. Left panel is
based solely on all-sky calibrations; right panel includes small
adjustments to V-mag based on 5 observing sessions and persistent
departures from a Star Color Sensitivity fit. I recommend use of
the right panel magnitudes.
Two stars are "dangerously" red for use with a slightly blue
Observing Start/Stop Local
If an observation of at least 1/2 hour is needed for a usable
measurement, and if measurements should be made when elevation (EL)
is above 20 or 30 degrees, then the following graphs can serve as a
guide for the season that KIC 846 can be monitored from the ground
under dark sky conditions (for my HAO observing site).
Local time for observing start and end, for the HAO observing
site (Latitude +31 deg, Longitude 110 E) for the requirement that
EL > 30 deg. Times for sunset + 1 hour and sunrise - 1 hour are
shown as dashed traces. Local midnight is 12:21 AM due to the
observing site's longitude being -110 deg (vs the standard -105
deg for the HAO time zone).
The number of hours that are potentially available for observing
is plotted in the next graph.
Number of hours when KIC846 is above specified elevation (EL),
and when the sky is dark (more than 1 hour past sunset and 1 hour
before sunrise) versus month number, for an observing site at
latitude +31 deg (Hereford Arizona Observatory).
From this graph it can be seen that January is the only month
when KIC846 can't be observed from ground observing sites for
monitoring purposes. December is marginal, since KIC846 is above EL
= 30 deg during darkness for only 0.6 hour (Dec 15). For observing
sites at more northerly latitudes the observing times will be
Boyajian et al, 2015, MNRAS, "Planet
Hunters X. KIC 8462852 - Where's the flux?"
Landolt, A., 2009, Astron. J., 137,
Montet, Benjamin T. and Joshua D. Simon, 2016,
"KIC 8462852 Faded Throughout the Kepler Mission," arXiv
Smith et al, 2002, Astron. J., 123,
Boyajian et al, 2015, MNRAS, "Planet
Hunters X. KIC 8462852 - Where's the flux?" link
Washington Post article, 2015.10.15: link
Notice requesting KIC646 observations
AAVSO LC Generator https://www.aavso.org/data/lcg
(enter KIC 8462852)
Web page tutorial: Tips for amateurs observating faint
asteroids (useful for any photometry observing)
Book: Exoplanet Observing for Amateurs,
Gary (2014): link
(useful for any photometry observing)
My web pages master list, resume
WebMaster: B. Gary. Nothing on this web page is copyrighted. This
site opened: Oct 15, 2015